Hart Splitter is used to splitting the signals. We need to configure the hart splitter according to our need.
P&F hart splitter KFD2-HLC-EX1.D.4S uses software viz Pactware for configuration.We can download the software free freom the following link: http://www.pepperl-fuchs.com/global/en/classid_163.htm?view=productlist&itemsperpage=50
The connection details is as follows:
- Connect the power supply (24v DC) across the terminals 23+ & 24-
- Connect the ABB Tx at the terminal (2+ 3- ) of HLC
- Note that the jumper is connected to HLC across the Terminals 4 & 5.
- Connect the HLC model to PC using K-ADP cable to configure using PACTWARE.
- The Measured value can be observed on PACTWARE.
- Output Current can be measured across terminals 8+ & 7- ,14+ & 13- , 20+ & 19-.